0 OBJECTIVE To lay down the calibration procedure of Magnetic stirrer to provide a system to calibrate control and maintain the equipment used for inspection measuring and test equipment in order to ensure the correctness of measurement and test status.
5 Startup procedure 9 6 Stirring operation 10 63 Recommended rotation speed ranges 13 6Operating conditions Magnetic stirrer 10 to 56 176C at max100 RH Power supply 5 to 40 176C at max.
1 Ensure that the instrument is clean2 Put the power plug in the socket and switch it ON3 Place the beaker or flask containing sample in the mid of Magnetic stirrer4 Put the magnet in the beaker or flaskcontaining sample5 Switch ON the stirrer and confirm ON position by power indicator LED lights up when the stirrer powered.
Connect the main supply of the instrumentSwitch ‘on’ the instrumentIf required adjust the required temperature by temperature ‘knob’Keep the beaker Having Sample on the plate and wait for heatingDeep the magnetic stirrer in a beaker and start the stirring by a knobStirring speed should be controlled by a ‘knob’ to slow middle and high.
Dec 28 2022nbsp01833251 Ensure that the instrument is clean and free from any dust particle2 Connect the main supply of the instrument3 Switch ‘on’ the instrument4 If required adjust the required temperature by temperature ‘knob’5 Keep the beaker Having Sample on the plate and wait for heating.
Jul 27 2008nbsp01833251 Ensure that the instrument is connected to the power supply2 Place the beaker containing the solution to be stirred in the center of the top plate and immerse the Teflon paddle into the solution3 Switch on the instrument the pilot lamp gets illuminated.
Jun 29 2020nbsp018332Operating procedure of Mechanical Stirrer Affix ‘Under Process’ label on the Mechanical Stirrer equipmentSwitch ‘ON’ the main of the Mechanical StirrerEnsure the proper fix of the dismantled stirrer shaft into the Mechanical StirrerPlace the container in which solution to be prepared below the Mechanical Stirrer.
Keep the shaft into the suspension to the half of the depth of suspension or up to that level so that suspension should not flush out4 Put ON the electric supply and start the stirring5 Stirring is carried out until a uniform slurry is obtained or time specified in BMR6 Switch OFF the mains after completion of the operation.
Magnetic Stirrer Operating Manual 50108226 Issue 022007 TM MINI 07 MINI 20 MICRO 07 MICRO 2016 Magnetic stirrer outlet 17 RS 232C port Figure 4 Control unit TELEMODUL 20 C left and TELEMODUL 40 C right.
Put on safety goggles and a pair of nitrile gloves for the remainder of this procedureTurn on magnetic stirrer hot plate to 35176 CIf your hot plate does not have an exact temperature setting then use an infrared or probe thermometerOperating Environment Recommendations for Best Performance Replacing the Particulate Carbon Filter.
Put on safety goggles an air respirator and a pair of Nitrile gloves for the remainder of this procedureTurn on your magnetic stirrer hot plate to 35176 CelsiusIf your hot plate does not have an exact temperature setting then use a infrared thermometerAdjust the heat dial until 35176 Celsius is achieved.
The magnetic stirrers is designed to handle the following stirring volume MONO DIRECT 1 to 3000 ml The drive and electronic control unit are sealed into the housing of the magnetic stirrer 2The housing is completely sealedThe magnetic stirrer 2 is cleared for use in safety laboratories and sterile rooms.
Thermo Scientific Magnetic Stirrer 31 Section 3 Operation To start the Stirrer follow the procedure below1 Pour the liquids to be mixed into the beaker or flask2 Place a magnetic stirring bar at the center of the beaker base3 Place a beaker on top of the setup plate or plate cover4 Turn the knob clockwise to start stirring.
To lay down the procedure for Operation of Stirrer0 SCOPE To lay down the procedure for Operation of Stirrer2 Checking Production Pharmacist amp Above 40 ACCOUNTABILITY HODProduction Assigned Designee 51 Production person shall ensure the cleanliness of the stirrer its part.
Copyright © 2021 Zilong Laboratory Machinery Company All rights reserved sitemap